Then, the Fixation Lever must be locked to ensure proper fixation. The following description of the surgical technique is using a Gamma3 Long Nail R1.
The corresponding nail lengths are marked on the Distal Targeting Arm Fig. Keep the Adjusting Device in its position as calibrated. Be sure that the Distal Targeting Arm is positioned anteriorly to the chosen nail If not, turn the Adjusting Screw until passage Open the catalogue to page 4 Introduction Distal Targeting System Components The major components of the device are made of carbon fiber material, providing radiolucency under the C-Arm imaging and stiffness for the distal locking procedure.
It is important to perform Distal Targeting Device calibration prior to the nail insertion.
For full Gamma3 Long Nail R1. Now make final adjustments with the drill, passing it through the most proximal hole in the nail as shown. The Fixation Lever must then be securely locked Fig. Compatibility Using the Distal Targeting System Then, look through the Tissue Protection Sleeve and adjust the targeting position by turning the Adjusting Screw until the holes of the sleeve and the nail appear coaxial.
Without using the Distal Targeting System, the placement of the distal locking screws is done primarily by a variety of freehand techniques, using conventional or radiolucent drilling devices.
These methods may result in repeated drilling, repeated X-Ray adjustment, which may require longer exposure as well as higher potential for mis-drilling2.
When this chapter is completed, the Set Screw has been properly positioned in a groove of the Lag Screw Fig. The white line must be visible in the Alignment Indicator Window Fig.
Insert the Fixation Bolt completely Fig. The drill must go through the nail hole smoothly and easily. This is required to secure the arm to the targeter and stabilize the system.
Open the catalogue to page 7 Operative Technique Distal Targeting Device Calibration The length of the chosen nail determines where the Adjusting Device should be attached. Place the assembled device onto this by placing it onto the metal pin as shown Fig. The white line must be seen through the Alignment Indicator Window for correct assembly Fig.
Release the Fixation Lever Fig.Stryker Osteosynthesis devices and instruments. It offers guidance that you should heed, but, as with any such technical Long Nail System and the T2 Recon Nail System. Distal Targeting System for the Gamma3 Long Nails and T2 Recon Nails Introduction Trocar, Long Alignment Indicator Window Fixation Bolt Fixation Lever.
Stryker Osteosynthesis devices and instruments. It offers guidance that you should heed, but, as with any such technical Besides the T2 Femoral nail with a 3m radius of curvature, Stryker offers also a m radius T2 Femoral Nail to complete the product offering for those patients with a higher anterior.
Stryker Osteosynthesis devices and instruments. It offers guidance that you should heed, but, as with any such technical The T2 Recon nail is one of the first femoral nailing systems to offer a greater trochanter entry point with both recon and antegrade locking options.
Take me to our location in. United States.
See additional locations. Recon Nailing System Operative Technique Trauma. Contributing Surgeons: Kyle F. Dickson, M.
D. using Stryker Trauma devices and instruments. It offers guidance that you should heed, but, as with any such technical T2™ Recon Nail for the treatment of complex, as well as more common fractures. Introduction Distal Targeting System for Gamma3 Long Nails and T2 Recon Nails In response to the request of surgeons around the world, Stryker Osteosynthesis has created a .Download